When trigger input switched to Ext, the scope is always triggering as it would in None mode. This bug looks to be implemented in one of the newest versions, it does not show up in 184.108.40.206 (Beta).
Search found 3 matches
Search found 3 matches • Page 1 of 1
- Thu Oct 14, 2010 2:47 pm
- Forum: PicoScope 6 for Windows
- Topic: AWG Sweep Sync - 5204
- Replies: 3
- Views: 2021
Well, Let's say I want to measure a filter V/F characteristics. Thus, I'm applying a sweep from the AWG to the input of the circuit, and I'm measuring the output with the CH1. To see a "nice" picture, the scope needs to get synced from the sweep "start" point, so that the full output fits the scope ...