Search found 4 matches
Search found 4 matches • Page 1 of 1
Hello Hans, I agree with Gerry. But at least for the PS2000 SDK, one can program the AWG waveform directly, should you be into programming. The function is ps2000_set_sig_gen_arbitrary(). Maximum length of the AWG waveform is 4096 samples per waveform cycle. This will allow you to go very much highe...
- Fri Mar 15, 2019 7:57 pm
- Forum: Pre-Sales Advice
- Topic: Limitations of PicoScope Software using PS 2205A
- Replies: 1
- Views: 1367
Hello, Having just acquired and explored a 2205A, I can confirm that the zoom window works well, if I understand your question properly. Test conditions were a 2 Hz sawtooth (is that what you mean by low?) from the integrated signal generator, 12 bits enhanced vertical resolution (by averaging), sam...
- Mon Mar 11, 2019 3:59 pm
- Forum: General Software Discussions
- Topic: Pre-trigger samples from ps2000_run_streaming_ns
- Replies: 1
- Views: 836
Hello, I am finding that the function ps2000_run_streaming_ns adds unwanted pre-trigger samples to the output stream. If I call the function with triggering disabled, and ask for a million samples, I receive exactly a million samples, which is fine. But if I enable simple triggering with the ps2000_...
- Mon Mar 11, 2019 2:17 pm
- Forum: .Net Languages (C#, VB.Net, .)
- Topic: PS2000 ps2000_overview_buffer_status return value
- Replies: 4
- Views: 3443
I am yet another who has now been caught out by the inverted return logic of the ps2000_overview_buffer_status function, or rather by its documentation. Can I point out that the 2000 series programmer's guide, that I downloaded last week as part of the 64-bit SDK, is still in error. It was only when...