thanks for your reply. So for a "real" calibration it is necessary to use the offered calibration service ?
Search found 6 matches
Search found 6 matches • Page 1 of 1
- Wed Jun 07, 2017 5:35 am
- Forum: USB PC Oscilloscopes
- Topic: PicoScope 2206b ADC
- Replies: 1
- Views: 1653
Hello Guys, I have 2 questions: 1. Are there any further on information's about the used ADC inside of the Picoscope 2206b ? 2. In the manual, there's a part about the Input Voltage areas. The Voltage areas are as follow : ±20 mV, ±50 mV, ±100 mV, ±200 mV, ±500 mV, ±1 V, ±2 V, ±5 V, ±10 V, ±20 V Doe...
Hey, I'm using a PicoScope 2206b and I want access it via LabVIEW. My PC runs on a 64 bit System and my LabVIEW is a 32 bit version. I Installed the 32 bit SDK (is this the right one for my case ?) and tried to start the Picoscope2000 32 bit examples from GitHub. I receive several Errors/ Missing VI...
- Wed Apr 12, 2017 8:05 am
- Forum: LabVIEW
- Topic: Error Message by trying the PS2000 Block example
- Replies: 1
- Views: 1065
Hey Guys, I'm using a PicoScope 2206b, the SDK and the software are installed. I wanted to start with the PS2000.vi example from the "PicoScope2000ExampleBox". So every time I try to run the example I receive following error: An error occurred while trying to initialize the unit. Attempt to disconne...