Search found 3 matches
Search found 3 matches • Page 1 of 1
Hi. I just received my 2205 MSO scope, but I have some problems that some NOT connected MSO channels are toggling high. For example, I've connected a 3.3 V, 10 kHz clock pulse on D0 and ground. I measure the same clock pulse with analog CHA. This is as expected. But I also see some very short "HIGH ...
- Sat Oct 01, 2016 12:22 am
- Forum: USB PC Oscilloscopes
- Topic: 50 ohm BMC attenuator
- Replies: 3
- Views: 1458
My Picoscope 2000 input range is +/- 20V, and probe in 10x mode is approved for 600 V pk-pk. Is it possible to connect a BMC attenuator to be able to measure up to 600 V? Is it a 50 ohm cable?