That solved my problem. Thank you very much!
Search found 5 matches
Search found 5 matches • Page 1 of 1
Hi, I've already fired off a couple of questions on the LabView forum, so I thought I should introduce myself before it's too late. My name is Alan. I'm working in the UK for a company designing pretty cool radiation detectors. Previously, I worked at CERN building detectors for the CMS experiment. ...
Hello Karunen. Thanks for the reply. Can you please clarify the steps? I always use the scope in 4 channel, 14-bit mode. Therefore, the ns/S equation is: ns/S = (T-2)/125,000,000. What exactly does my while loop converge upon? I've been reading the paragraph in the manual over and over on this topic...
- Thu Feb 19, 2015 2:47 pm
- Forum: LabVIEW
- Topic: Getting triggered info in block mode
- Replies: 2
- Views: 1654
Hi all, I have written a LabView GUI using block mode which works well, except that I want to add a counter of the number of triggered events. I see this information is available when using the Streaming mode, but I don't see how to achieve it in block mode. How can I obtain the status of the trigge...
Hi all, I have a Picoscope 5444B which I am almost done writing a LabView program for. I have a serious time constraint so I am asking here for help to sort out my last issue. I need to get the timebase set up correctly so that I can display the time/div just like a conventional scope. I'm having tr...