If you have installed the Instrument Driver files via the MATLAB Add-Ons Explorer, then please update the Instrument Driver files as a new property, displayOutput, has been added.
To suppress the output to the command window during data collection, please add the following to the MATLAB script after the connect code e.g.
If you wish to wait until after the data has been retrieved before converting to millivolts then you will need to call the ps3000aSetDataBuffer() function to set a buffer for each channel/segment combination and ps3000aGetValuesBulk() function in order to retrieve the data instead of using the getRapidBlockData() function.
Code: Select all
connect(ps3000aDeviceObj); set(ps3000aDeviceObj, 'displayOutput', 0);
When installing the files on another PC, I would recommend the following:
- Install the PicoSDK (32-bit/64-bit according to your version of MATLAB)
- Use the MATLAB Add-Ons Explorer to install the PicoScope 3000 Series - MATLAB Generic Instrument Driver and PicoScope Support Toolbox
I hope this helps,