The PicoScope 9200 and 9300 Series are leadingedge product families resulting from the same long and continuing program of product development. The PicoScope 9300 Series shares many features with the 9200 Series but has updated hardware and software. In this comparison of the two families we will go beyond the headline specifications to consider which model is suited to your application. The discussion includes the recently introduced PicoSource PG900 fast pulse generators, which use the same fast pulse technology found in the 9300 Series sampling oscilloscopes.
Leaving aside parameter specifications for a moment, you first need to decide on the hardware configuration—the combination of input and output channels—that best suits your requirements.
As shown in Figure 1, the range of models that we offer is extensive. The 12 GHz PicoScope 9200 Series and the 20 GHz PicoScope 9300 Series are both available with and without TDR/TDT sources, opticaltoelectrical converters and clock recovery capability. The PicoSource PG900 Series is available with builtin 60 ps outputs, external 40 ps outputs or both together.
Requirement  PicoScope 9200  PicoScope 9300  PicoScope 9300 + PicoSource PG900 

Twochannel sampling oscilloscope  9201  9301  
Fourchannel sampling oscilloscope Often needed in differential applications 
9341  9341  
Time domain network, cable, port or component analysis 
9211  9311 or 9312  Any 9300 
Differential TDT or TDR  9311 or 9312  Any 9300  
Optical fibre applications  9231  9321  9321 
Recovery of clock from the data in serial data applications  9211 or 9231  9302 or 9321  9302 or 9321 
Ethernet LAN control connection  9211 or 9231  All 9300 
All 9300 
Approximate price range  $10k  $25k  $15k  $30k  $23k  $43k 
Warranty  Now 5 years for all models and configurations 
Table 1 – Selection of model or configuration
The bandwidth of your oscilloscope must be large enough for the signal speed or bandwidth (related to its frequency, data rate or transition time). Higherbandwidth instruments also have faster triggers, shorter sampling interval (time resolution) and lower timing jitter. These parameters may determine your choice of oscilloscope family.
Parameter  PicoScope 9200 Series  PicoScope 9300 Series 

Channel input –3 dB bandwidth ^{[1]}  12 GHz  20 GHz 
Channel input rise/fall time^{[1]}  29 ps  17.5 ps 
Data rate that can be viewed, max.^{[1]}  24 Gb/s  40 Gb/s 
Data interval or pulse width that can be viewed, min.^{[1]}  42 ps  25 ps 
Data rate that can be characterized to 3rd harmonic, max.  8 Gb/s  13.3 Gb/s 
Data interval that can be characterized to 3rd harmonic, max.  125 ps  75 ps 
Data rate that can be characterized to 5th harmonic, max.  4.8 Gb/s  8 Gb/s 
Data interval that can be characterized to 5th harmonic, max.  208 ps  125 ps 
Sampling interval (time resolution), min.  0.2 ps  0.064 ps 
Effective sampling rate, max.  5 TSa/s  15 TSa/s 
Sampling jitter (time smear due to the oscilloscope in the displayed waveform or eye) 
4 ps RMS  2.0 ps RMS 
Microwave bands covered^{[1]}  I, G, P, L, S, C, X.  I, G, P, L, S, C, X, K. 
_{Note [1]: See also maximum trigger rate constraints below}
Table 2 – Selection of input bandwidth and sampling parameters
A peculiarity of sampling oscilloscopes is that trigger bandwidth rarely matches channel or sampler bandwidth. To facilitate stable trigger (and hence waveform display) from very high signal frequencies, all PicoScope 9200 and 9300 sampling oscilloscopes feature a separate prescaled trigger input. Here the trigger rate is divided down to a rate that the sampling oscilloscope can accept. This is sufficient on a sampling oscilloscope as a repetitive signal and trigger are required anyway. You should therefore always consider prescaler bandwidth when selecting a product family, noting that a further externally divided or subharmonic trigger signal will be required to view a stable waveform at higher frequencies than those listed in the table above.
Parameter  PicoScope 9200 Series  PicoScope 9300 Series 

Trigger frequency (max.) for stable direct trigger  1 GHz  2.5 GHz 
Clock period (min.) for stable direct trigger  1 ns  400 ps 
Data rate (max.) for stable direct trigger  2 Gb/s  5 Gb/s 
Trigger prescaler frequency (max.) for stable trigger  10 GHz  14 GHz 
Trigger prescaler clock period (min.) for stable trigger  100 ps  71 ps 
Trigger prescaler data rate (max.) for stable trigger  20 Gb/s  28 Gb/s 
Table 3 – Selection of trigger and prescaler bandwidth
When working with data streams, the data clock is often not available. This is particularly common when working with serial data from an optical fiber, but also with many highspeed electrical dataonly streams where clock recovery is used to eliminate clocktodata skew or accumulated jitter. It is therefore usual to find a clockrecovered trigger capability provided with broadband oscilloscopes, and in the PicoScope 9000 Series this is offered on the models below.
PicoScope 9211 / 9231  PicoScope 9302 / 9321  

Clock recovery data rate  12.3 Mb/s to 2.7 Gbs/s  6.5 Mb/s to 11.3 Gb/s 
Clock recovery frequency  6.15 MHz to 1.35 GHz  3.25 MHz to 5.65 GHz 
Recovered clock trigger jitter  < 1 ps + 1% of unit interval RMS  < 1.5 ps RMS 
Table 4 – Clock recovered trigger
When considering serial data applications, you may need to consider pattern sync (or trigger divide by n) capability. By counting and subdividing trigger rate, a sampling oscilloscope can synchronize to and create a stable display of a serial data pattern. You can then search the whole pattern for data or eye corruption using the eye line feature. The maximum rate and count length over which this can be achieved may also be important. Pattern sync is available on all PicoScope 9200 and 9300 models
PicoScope 9200 Series  PicoScope 9300 Series  

Pattern sync (divide by n) trigger rate  10 Mb/s to 8 Gb/s  10 Mb/s to 11.3 Gb/s 
Pattern sync (divide by n) length  7 to 2^{16}–1 (65 535)  7 to 2^{23}–1 (8 388 607) 
Table 5 – Pattern sync trigger
A sampling oscilloscope does not have the switched voltage ranges seen on a realtime oscilloscope. Instead it has a single fixed voltage range and a highresolution analogtodigital converter (16bit in the PicoScope 9200 and 9300 Series). Fullscale range is ±1 V DC or AC peak. The converted signal is digitally scaled to achieve all other voltage ranges. Note that even at ±4 mV (1 mV/div), 8 ADC bits remain available.
A sampling oscilloscope (in its sequential sampling mode) gathers only one trace sample for each trigger event. It is therefore essential that it triggers, stores and rearms as fast as possible. In this respect the 9200 Series is typical amongst sampling oscilloscopes, while the PicoScope 9300 Series with its ARM processor and SPARTAN6 FPGA stands out as one of the fastest available. Traces, eyes, graded persistence displays, measurements and statistical assessments are all as fast as they come on the 9300 Series!
Thanks to the faster sampling in the PicoScope 9300 Series, trace length can realistically be increased to an industryleading 32k samples. This improves view and zoom detail, measurement resolution and resolution of mathematical functions such as FFT, and more closely matches the available vertical and horizontal resolutions. Additionally, both series can take full advantage of a connected HD display size and resolution and can even stretch this waveform and persisted display detail (16 bits and 32k samples) across multiple monitors.
A sampling oscilloscope has no input amplification. Despite this, noise is still present in the system and limits the smallest signals that are viewable. The PicoScope 9300 Series front end is quieter than the 9200 Series but has greater bandwidth. In practice, the minimum viewable signal level is similar at similar degrees of averaging. However, the faster PicoScope 9300 Series can average more deeply while achieving a similar update rate.
PicoScope 9200  PicoScope 9300  

ADC resolution 
16 bits or <40 µV / bit  16 bits or <40 µV / bit 
Input RMS noise  <2 mV (58 nV/√Hz)  <2 mV (45 nV/√Hz) 
Input RMS noise with best averaging  <0.1 mV (2.9 nV/√Hz)  <0.1 mV (2.2 nV/√Hz) 
ADC sampling rate  200 kSa/s  1 MS/s 
Trace length  4 kS  32 kS 
Table 6 – Comparison of ADC resolution, noise, speed and trace length
The bandwidth of the opticaltoelectrical converter (and any trigger or clock recovery pickoff applied) limits the bandwidth and data rate of any optically derived signal, as can be seen in the table below. Maximum clock recovery rate, pattern sync and trigger rate and their pickoff bandwidths may also need to be considered.
A further consideration is that opticaltoelectrical receivers (and their transmitter counterparts) tend to distort the data eye, mainly because of their asymmetric transition times. Receivers, including the PicoScope 9000 Series sampling oscilloscopes, are provided with electrical lowpass filters to correct the eye at particular data rates. Typically these have a –3 dB cutoff of 0.75 x data rate, and oscilloscope bandwidth needs to be higher, about equal to the data rate. Please contact Pico Technology technical support if a particular data rate does not appear to be covered by the accessory filters listed for these products.
You should also check that the opticaltoelectrical converter, common to the 9200 and 9300 Series, is compatible with the fiber mode and connector (singlemode, multimode, FC/PC), peak power +7 dBm and carrier wavelength of the application to be addressed. The wavelength ranges are 750 nm to 1650 nm, spot calibrated at 850 nm (MM), 1310 nm (MM/SM), 1550 nm (SM).
PicoScope 9231  PicoScope 9321  

Optical bandwidth via the optical to electrical converter  8.5 GHz  9.5 GHz 
Bandwidth including trigger pickoff using supplied power divider  4 GHz  9GHz 
Data rate, max., for viewing See also clock recovery rate and pattern sync ranges 
17 Gb/s  19 Gb/s 
Data rate, max., for viewing, using supplied power divider  8 Gb/s  18 Gb/s 
Data rate, max., for characterization  12 Gb/s  19 Gb/s 
Data rate, max., for characterization, using supplied power divider  8 Gb/s  18 Gb/s 
_{* including trigger pickoff using supplied power divider (faster divider available)}
Table 7 – Comparison of optical signal bandwidth and data rates
Time domain transmission and reflectometry both apply a fastedge pulse to an unknown single or multiport network. This could be an electrical component, a cable, or a signal receive or transmit port. By measuring the pulse that reflects back (TDR) or the pulse that passes through (TDT), the transmission characteristic or mismatch can be analyzed.
Scalar or vector network analyzers perform a similar function to TDR/TDT analyzers but use a frequencydomain instead of a timedomain technique. These instruments inject and measure a sine wave, one frequency at a time. TDR/TDT instruments inject a fast pulse containing a wide spectrum of frequencies all at once.
TDR/TDT analysis tends to be faster, while scalar or vector analysis tends to have more dynamic range. Results, for essentially the same measurement, can be converted between the two domains. Timedomain measurements are particularly effective at determining the physical location of a mismatch, fault or component. Physical distance is obtained simply by multiplying the timing of a reflection by its propagation velocity (close to the speed of light, c). Vector quantities such as sparameters can theoretically be extracted from both techniques, but presently the PicoScope 9200 and 9300 Series extract only scalar quantities.
System bandwidth or, more accurately, system transition time (that of the fast pulse, combined with that of the sampling oscilloscope and any interconnect) determines the time resolution and hence the physical distance that can be resolved. To resolve the magnitude of a mismatch, a complete rise time and fall time are needed to cleanly define a pulse amplitude. When detecting the presence or location of a mismatch, distance resolution can be around five times better than this. Resolution quantities given below assume typical PCB or coax propagation velocity of around 0.7 c.
Available pulse amplitude and variability of pulse amplitude can both be considerations, but not as significant as you might think. Usable pulse amplitude is limited in practice to the peak fullscale voltage that the oscilloscope can accept: in these instruments, 1 V pk from an opencircuit reflection or unimpeded transmission path. The benefit of large amplitude at the source (as with the PicoScope 9311) is that the test system source match can be improved when using an attenuator. Variable amplitude gives the opportunity to optimize signal level and thus dynamic range in any TDT/TDR measurement, and possibly reduce amplitude to keep within its deviceundertest limits or, in the case of an active device, explore its nonlinearity.
Pulse distortions or aberrations at the pulse source or at the receiving oscilloscope and interconnect are not of primary importance in a TDT/TDR application because they are corrected by system port calibration into known short, open and load.
All the PicoScope TDR/TDT solutions include two generators and two receive input channels. Two TDR or TDT measurements can be performed simultaneously in each case. However, the PicoScope 9200 Series and PicoScope 9000 software are limited to singleended measurements, while the PicoScope 9300 series and PicoSample 3 software can address true differential, singleended and commonmode measurements for transmission lines or ports with or without a physical ground.
A further benefit of the PicoScope 9300 series is the option of either variable and highamplitude pulse generators, based upon step recovery diode technology, or slightly faster, but fixed and loweramplitude generators, based on tunnel diode pulse heads. In general, the step recovery diode offers more flexibility.
As a third configuration option, the TDR/TDT generator can be separated from the oscilloscope. There is a choice of three generators from the PicoSource PG900 family with either SRD or TD generation, or both. These offer the flexibility of separately located transmit and receive TDT as well as standalone use. The PicoSource PG900 Series generators are compatible with all models in the PicoScope 9300 series, allowing for instance the combination of TDR/TDT with optical, clock recovery or fourchannel capability.
TDR/TDT functionality is substantially enhanced in the PicoSample 3 software supplied with the PicoScope 9300 Series. As noted in the table above, differential TDR/TDT is supported, with full correction for differential coaxial (twinaxial) cable, and for parallel or twisted lines where no ground is present.
PicoScope 9211 or 9231  PicoScope 9311 or 9312  PicoScope 9300 + PicoSource PG900  

Singleended (normal mode) TDR/TDT 
Yes, dual channel  
Differential mode TDR/TDT 
No  Yes  Yes 
System 10% to 90% transition time 
500 ps  60 ps  60 ps 
System 10% to 90% transition time with PicoScope 9312, PicoSource PG912 or PicoSource PG914 TD heads 
N.A.  40 ps  40 ps 
Best system transition time after correction of system aberrations 
100 ps  40 ps  40 ps 
Typical minimum length for accurate impedance measurement 
40 mm  16 mm  16 mm 
Typical best resolution of fault or distance to fault 
8 mm  4 mm  4 mm 
Maximum TDR / TDT analysis period  600 ns  8 us  8 us 
Typical maximum distance to fault in TDR  60 m (200 ft)^{*}  400 m (1350 ft)^{**}  400 m (1350 ft)^{**} 
TDT/TDR pulse amplitude 
400 mV  2.5 to 6 V  2.5 to 6 V 
TDT/TDR pulse amplitude with TD pulse head 
N.A.  200 mV  200 mV 
_{* Assuming propagation velocity of 70% c (speed of light in a vacuum) and total path length attenuation < 20dB @ 100 MHz}
_{**Assuming propagation velocity of 70% c (speed of light in a vacuum) and total path length attenuation < 20dB @ 10 MHz}
Table 8 – Time domain network (TDR/TDT) analysis capability
The PicoScope 9200 and 9300 Series oscilloscopes are supplied with different software. PicoSample 3, supplied with the 9300 Series scopes, is a modern, graphically enhanced, wideformat and touchscreenenabled evolution of its predecessor, the PicoScope 9000 software supplied with the PicoScope 9200 Series. User interface and key feature enhancements in PicoSample 3 are highlighted in the comparison below. You can download both applications and use them in demonstration mode to explore the differences.
PicoScope 9200 Series and PicoScope 9000 software  PicoScope 9300 Series and PicoSample 3 software  

General appearance of fully resizable Windows user interfaces  Improved layout for widescreen monitors 

Select left and right menus according to application 
Controls, traces and measurements areas can be configured to application  Select any two menus or dismiss them to focus on traces and results 
Radio buttons, check boxes and submenus 
Selection, action and submenu controls  Graphic icons, dropdowns and tabs 
Increment and keyboard entry 
Parameter controls  Innovative course, fine, defaults, slidebar, keyboard and calculator entry in a single control. 
Mouse click and limited drag function. No specific touchscreen enhancements. 
Touchscreen support  Click, drag, enlarged controls and calculator entry. Full touchscreen operation. 
Enhanced functions  Math and FFT functions, display modes, trace labels, differential TDR/TDT 
The 9300 series are slightly longer and slightly heavier than the 9200 series.
The 9200 series dimensions are 170 mm wide x 255 mm length x 40 mm high (6.7 x 10.0 x 1.6 in) and weight 1.1 kg (2.4 lb) max.
The 9300 series are 170 mm wide x 285 mm length x 40 mm high and 1.2 kg (2.6 lb) max.