The PicoScope 9000 Series 12 GHz PC Sampling Oscilloscopes are specialized instruments designed for analyzing serial data signal quality; for characterizing cables, connectors and circuit boards; and for TDR/TDT testing. 8 GHz optical-input models are also available. These scopes require careful setting up to obtain accurate results, so we have produced a series of instructional videos to help you use their capabilities to the full.
Here are the latest video releases:
Watch out for more videos later!
Explore our library of videos on our YouTube channel
Forum contributor FrancisL has posted a report on his measurements using a DrDAQ to evaluate the effect of cavity wall insulation. He has used the original parallel-port DrDAQ, but the experiment can easily be upgraded to use the new USB DrDAQ, which has all the features of the original device and many more:
In July Pico Technology will celebrate its 20th anniversary! As part of the festivities, we plan to publish stories of the oldest Pico products still in use. If you are using an early Pico oscilloscope or data logger, please let us know which one it is and what you are using it for. We will publish the most interesting applications in our July newsletter.
To prevent signal distortion, it is important that attenuated oscilloscope probes are correctly compensated or ‘tuned’ to match the oscilloscope. We can supply factory-tuned passive probes for use with our higher-bandwidth scopes such as the PicoScope 3000A, 3000B and 5000 Series. If you already own a general-purpose probe like the 250 MHz MI103, this application note will show you how to tune it for use with any oscilloscope. The article covers both low-frequency and high-frequency compensation, but you can omit the HF adjustment if your probe has only LF compensation.
Here are some of the questions and answers that have cropped up recently on our discussion forum:
Q. Connecting two HumidiProbe sensors. In the description page of the HumidiProbe sensors it is mentioned that up to 4 can be connected to a PC running PicoLog. However, I cannot find a way to have my two sensors monitored at the same time: if I go to “Settings > Input channels…” I see both my sensors but I can only choose one of them. Is there something I am missing?
A. When you go into “File > New Settings”, the “Recording” dialog will open. In the bottom left-hand corner there is a tick box labelled “Use Multiple Converters”. If you tick that box, you will be able to select more than one converter in the following dialogs. PicoLog now supports up to 20 devices at once.
Q. Channel markers in PicoLog. When I first used PicoLog, different-coloured markers appeared on the graphs. Now I no longer see them, even after restarting the program and starting new recordings. I have found the box for “Display markers” and this is ticked. What else can I try?
A. The “All traces on one graph” format (Graph View > Options Button > Format) does not automatically switch on the markers. Change the format to “Automatic” and the markers will return.
Please visit Pico Exhibitions for the latest list of exhibitions and trade shows that Pico and its representatives will be attending.
Thanks to our continuing success and growth Pico are always seeking talented people to join our company.
Please visit https://jobs.picotech.com/ to see our current vacancies. We look forward to hearing from you!
Our latest software releases are available as free downloads. To check which release you are using, start the software and select Help > About.
To make sure that your Pico newsletter reaches your inbox every month, add email@example.com to your email address book or safe list. If you found this newsletter useful, please recommend it to your friends and colleagues. Back issues are available from our newsletter archive.
By subscribing to our newsletter, you agree to receive one email a month plus no more than two special announcements a year. We will not use your email address for any other purpose unless separately agreed with you.
Pico Technology, James House, Colmworth Business Park, St. Neots,
Cambridgeshire, PE19 8YP, United Kingdom
Tel.: 01480 396395 (+44 1480 396395)
Fax: 01480 396296 (+44 1480 396296)
North America Office:
Pico Technology, 320 N Glenwood Blvd, Tyler TX 75702, United States
Tel:+1 800 591 2796 (Toll Free)
Fax:+1 620 272 0981
Pico Technology, Room 2252, 22/F, Centro, 568 Hengfeng Road, Zhabei District,
Shanghai 200070, PR China
Tel: +86 21 2226-5152