## DrDAQ

#### Data logger, oscilloscope, signal generator and more!

DrDAQ is a versatile instrument that connects to the USB port of any PC. Using the supplied PicoScope software it can be used as an oscilloscope, spectrum analyzer and signal generator.

# Can your recorder tell you the speed of sound?

With the help of DrDAQ educational data logger and PicoScope’s Spectrum Analyser, see how you can determine the speed of sound using a recorder.

## Introduction

This Science experiment is adapted from the Salters Horners Advanced Physics Project. It corresponds to Activity 11 from the ‘Sound of Music’ section of the course.

A recorder is basically an open pipe which, when played in the normal way, develops a standing wave between the open end where you blow and the first open hole below the mouthpiece. The exact locations of the displacement antinodes at each end of the standing wave can be quite difficult to pinpoint but, for this experiment, it is sufficient to take them as the tip of the mouthpiece and the far end of the first open hole (Figure A11.1).

Figure A11.1: cutaway side view showing how to play note B

The relationship between the length l of an open tube and the frequency f of the longest standing wave that it could accommodate is given by the expression:

f = v/2l where v is the speed of sound in air.

The spectrum analyzer allows you to measure the frequencies in a recorded sound.

## Preparation

Measuring the frequency of just one note would give a crude value of the speed of sound. A much better technique would be to measure f and l for several different notes, and plot a graph.

## Questions

1. Sketch the shape of the graph you would expect if you plotted (a) f against l and (b) f against 1/ l.
2. What would be the gradient of a graph of f against 1/ l?

## Measurements and calculations

Draw up a table for recording your values of f, l and 1/ l. Take care to head each column with appropriate units.

Load the PicoScope software and bring the Spectrum Analyzer on screen. Set the Frequency Range to 1200 Hz; the X gain to 1; Input A to Sound; the Trigger to Single, Sound, Rising, 40 mV*; and the Display setting (right-hand box at the foot of the screen) to 0%. Click GO and play the note C as shown in Figure A11.2.

* You may be able to lower this value but it must be large enough to stop triggering too soon.

Figure A11.2: notes on a recorder

A frequency spectrum like that in Figure A11.3 should appear. Click the cursor on the peak of the lowest frequency present with a relatively large amplitude. In Figure A11.3 this is at 536 Hz. Record this frequency, its fundamental frequency, in your table of results.

Figure A11.3: frequency spectrum for note C

Click on ‘View’ and then ‘Clear’ in the drop-down menu that appears.

Repeat the process for the other notes shown in Figure A11.2, each time noting down the fundamental frequency.

Measure the approximate length l of the standing wave for each note. Record these lengths in the table.

Calculate 1/ l in each case and record these values in the table also.

Plot a graph of f against 1/ l, taking care with its presentation:

• Choose a sensible scale
• Label axes with their units.
• Use a sharp pencil to plot each point as a small cross or as a small dot plus circle.
• Give the graph a title.

Draw a line of best fit - a single straight line drawn with a ruler.

Draw a right-angled triangle, as large as possible, with the line of best fit as its hypotenuse.

Calculate the gradient of your graph and hence calculate the speed of sound in air.

## Further discussion

The speed of sound in air is about 330 m/s at 0 °C and about 340 m/s at 20 °C. Compare your results with these values.

What do you think was the largest source of inaccuracy in the measurements that you took?

Your line of best fit might not have passed exactly through the origin of your graph. How do you think this could best be explained?

## Testimonials

• The kit (2408B) is of obvious quality, easy to setup and calibrate and the free to download software has a reasonable learning curve. Superb kit, superb support, what more can I say.

Rop Honnor
• I have been using my 4224 PicoScope for years. I travel abroad so this has been ideal due to its physical size. Storage of waveforms on my Laptop is very easy allowing me to quickly email waveforms to my Colleagues.

Andrew
• Not many USB scopes works on Win & Mac & Linux too, so that proves me that guys from Pico really cares about us, customers. This made my decision much easier when I was looking to buy an USB scope.

Raul Trifan
• We have been using Picoscope 6404D for quite some time, and are amazed by its accuracy and powerful emulations while working with numerous signal evaluations.

J Mohanty
• PicoLog TC-08: This is a very nice unit that works consistently and reliably.

Jeff Hulett
• It is a great scope. I had a weird problem - it did not work on one of my PC’s. Customer service gave me first class service. If I could give 6 stars for customer service - I would do so.

Niels Larsen
• Perfect Partner for Development of Encoder controlled Stepper Motor Actuators. Since the included software is really stable, this type of device is a great tool for great tasks!

Helmut Schoettner
• A superb piece of equipment worth its weight in gold

Nigel Clinch
• So simple to use & beats any other I have ever used hands down.

John D Samsing
• Rather than webcam our benchtop scope screen, it was suggested we use a PicoScope to share waveforms via screen share.

We were a bit hesitant at first … but it worked flawlessly, and everybody is now out shopping for PicoScopes!

T Lövskog • SVP of Innovation at Additude AB
• Great functionality in a compact size. I really like moving the mouse pointer to a position and having the Time and Voltage display the values at that point. Calibration equipment is a breeze with that feature.

Don Horein

PC oscilloscope and data logger products