# PT100 platinum resistance thermometers

Platinum resistance thermometers (PRTs) offer excellent accuracy over a wide temperature range (from –200 to +850 °C). Standard sensors are are available from many manufacturers with various accuracy specifications and numerous packaging options to suit most applications. Unlike thermocouples, it is not necessary to use special cables to connect to the sensor.

The principle of operation is to measure the resistance of a platinum element. The most common type (PT100) has a resistance of 100 ohms at 0 °C and 138.4 ohms at 100 °C. There are also PT1000 sensors that have a resistance of 1000 ohms at 0 °C.

The relationship between temperature and resistance is approximately linear over a small temperature range: for example, if you assume that it is linear over the 0 to 100 °C range, the error at 50 °C is 0.4 °C. For precision measurement, it is necessary to linearise the resistance to give an accurate temperature. The most recent definition of the relationship between resistance and temperature is International Temperature Standard 90 (ITS-90).

The linearization equation is:

Rt = R0 * (1 + A* t + B*t2 + C*(t-100)* t3)

Where:

Rt is the resistance at temperature t, R0 is the resistance at 0 °C, and
A= 3.9083 E–3
B = –5.775 E–7
C = –4.183 E–12 (below 0 °C), or
C = 0 (above 0 °C)

For a PT100 sensor, a 1 °C temperature change will cause a 0.384 ohm change in resistance, so even a small error in measurement of the resistance (for example, the resistance of the wires leading to the sensor) can cause a large error in the measurement of the temperature. For precision work, sensors have four wires- two to carry the sense current, and two to measure the voltage across the sensor element. It is also possible to obtain three-wire sensors, although these operate on the (not necessarily valid) assumption that the resistance of each of the three wires is the same.

### Risk of self-heating

The current through the sensor will cause some heating: for example, a sense current of 1 mA through a 100 ohm resistor will generate 100 µW of heat. If the sensor element is unable to dissipate this heat, it will report an artificially high temperature. This effect can be reduced by either using a large sensor element, or by making sure that it is in good thermal contact with its environment.

Using a 1 mA sense current will give a signal of only 100 mV. Because the change in resistance for a degree celsius is very small, even a small error in the measurement of the voltage across the sensor will produce a large error in the temperature measurement. For example, a 100 µV voltage measurement error will give a 0.4 °C error in the temperature reading. Similarly, a 1 µA error in the sense current will give 0.4 °C temperature error.

Because of the low signal levels, it is important to keep any cables away from electric cables, motors, switchgear and other devices that may emit electrical noise. Using screened cable, with the screen grounded at one end, may help to reduce interference. When using long cables, it is necessary to check that the measuring equipment is capable of handling the resistance of the cables. Most equipment can cope with up to 100 ohms per core.

### Choosing PT100 sensors

The type of probe and cable should be chosen carefully to suit the application. The main issues are the temperature range and exposure to fluids (corrosive or conductive) or metals. Clearly, normal solder junctions on cables should not be used at temperatures above about 170 °C.

Sensor manufacturers offer a wide range of sensors that comply with BS1904 class B (DIN 43760): these sensors offer an accuracy of ±0.3 °C at 0 °C. For increased accuracy, BS1904 class A (±0.15 °C) or tenth-DIN sensors (±0.03 °C). Companies like Isotech can provide standards with 0.001 °C accuracy. Please note that these accuracy specifications relate to the SENSOR ONLY: it is necessary to add on any error in the measuring system as well.

Related standards are IEC751 and JISC1604-1989. IEC751 also defines the colour coding for PRT sensor cables: the one or two wires attached to one end of the sensor are red, and the one or two wires at the other end are white.

### Pico products for use with PT100 sensors

The Pico Technology PT-104 is a complete data logging solution for recording temperature using PT100 and PT1000 sensors offering the ultimate in accuracy (0.01 °C) and resolution (0.001 °C)

## Testimonials

• The kit (2408B) is of obvious quality, easy to setup and calibrate and the free to download software has a reasonable learning curve. Superb kit, superb support, what more can I say.

Rop Honnor
• I have been using my 4224 PicoScope for years. I travel abroad so this has been ideal due to its physical size. Storage of waveforms on my Laptop is very easy allowing me to quickly email waveforms to my Colleagues.

Andrew
• Not many USB scopes works on Win & Mac & Linux too, so that proves me that guys from Pico really cares about us, customers. This made my decision much easier when I was looking to buy an USB scope.

Raul Trifan
• We have been using Picoscope 6404D for quite some time, and are amazed by its accuracy and powerful emulations while working with numerous signal evaluations.

J Mohanty
• PicoLog TC-08: This is a very nice unit that works consistently and reliably.

Jeff Hulett
• It is a great scope. I had a weird problem - it did not work on one of my PC’s. Customer service gave me first class service. If I could give 6 stars for customer service - I would do so.

Niels Larsen
• Perfect Partner for Development of Encoder controlled Stepper Motor Actuators. Since the included software is really stable, this type of device is a great tool for great tasks!

Helmut Schoettner
• A superb piece of equipment worth its weight in gold

Nigel Clinch
• So simple to use & beats any other I have ever used hands down.

John D Samsing
• Rather than webcam our benchtop scope screen, it was suggested we use a PicoScope to share waveforms via screen share.

We were a bit hesitant at first … but it worked flawlessly, and everybody is now out shopping for PicoScopes!

T Lövskog • SVP of Innovation at Additude AB
• Great functionality in a compact size. I really like moving the mouse pointer to a position and having the Time and Voltage display the values at that point. Calibration equipment is a breeze with that feature.

Don Horein

PC oscilloscope and data logger products